Here you have full control over all the HV bias and amplifier settings.
- First, switch on the amplifier power,
- Now switch on the HV Bias power,
- Slowly ramp up the bias voltage to about 2500 - 3000V,
- You should now have an input signal with a low noise level,
the scrollbar on the left to bring the source closer to the detector,
- Use the other controls to observe their effect on the oscilloscope signal.
- Use the Run/Stop button to freeze the display.